Read Online or Download Agilent 35670A Analyzer (operators) PDF
Similar nonfiction_6 books
This guide applies to lively responsibility, reserve, nationwide protect, emergency crucial civilian, and emergency crucial agreement group of workers. It compiles conflict ability strategies, strategies, and approaches from many resources right into a pocket-sized, speedy reference consultant. This guide implements AFPD 10-25 Full-Spectrum risk reaction and comprises provisions of AFMAN 10-2602 Nuclear, organic, Chemical, and traditional (NBCC) protection Operations and criteria, in addition to parts of AFH 32-4014, Vol four, USAF skill to outlive and function techniques in a Nuclear, organic, and Chemical (NBC) atmosphere that's rescinded upon e-book of this handbook.
- Ordnance and Gunnery - USNI Chap 12 (gunsights) Part 1
- Flavonoids: Biosynthesis, Biological Effects and Dietary Sources
- The Admiralties - Opns, 1st Cav Div, [29 Feb - 18 May 1944]
- MOSELEY TRC-15A Remote Control System (broadcast radio)
- NIF - Assembly, Maintenance of Full-Scale [laser] Amplifiers
- Reactor and Materials Tech Monthly Rpt [May, 1965] [Declassified]
Extra info for Agilent 35670A Analyzer (operators)
5 Press [Scale], then press [AUTOSCALE ON OFF] to highlight ON. 6 Turn on the calibrator, then press [Start]. 7 After the display has been updated at least once, move the marker to the calibrator’s output frequency and enter calibrator’s output amplitude. Press [Marker] [MARKER TO PEAK]. Press [Input] [XDCR UNIT CHx SETUP] [CAL VALUE AT MARKER], then type the calibrator’s specified output amplitude and press [ENTER]. The analyzer uses the number you enter in [CAL VALUE AT MKR] to calculate a new value for [XDCR SENSITVTY].
This makes it easier to compare data on both traces, particularly when you have the upper/lower or front/back display formats selected. Axes scale markers The Agilent 35670A offers a feature called axes scaling. This lets you expand the displayed trace to examine it more closely. Using the axes scale markers, you can: • Specify X-axis or Y-axis scaling. • Move markers using the knob or numeric entry keys. • Return to a full-span display. • Expand the band identified by the markers to fill the display.
For some measurements, averaging lets you discriminate between random events (noise, for example) and components that are actually part of the signal you want to characterize. Part I-12 Agilent 35670A Operator's Guide Setting Up a Measurement Triggering Starting a measurement is a three-state process. The first stage is pressing the [ Start ] key, the second stage is arming the measurement, and the third stage is triggering. Arming merely enables the analyzer to respond to a trigger. Triggering tells the analyzer to begin collecting data.